Determination of pH in wine
With CDR WineLab Touch is possible to determine pH in wine with a colorimetric method whose results are correlated to the results of the OIV method which is based on an electrochemical measuring.
WineLab method, that does not require the use of any probe, does not need maintenance or calibration.
CDR WineLab Touch: analyses for wine and must
The system is supplied pre-calibrated and ready for use.
Results are expressed in accordance with the reference method.
It is also possible to standardize the system using samples with a known titration.
The reagents have a shelf-life of 12 months.
The analytical methods are simple: the system does not require an equipped laboratory or skilled staff.
Test type: End Point.
Colorimetric reaction at 630 nm.
Time testing: one test 1 minute - 80 Test/Hour.
Are possible test sessions with several samples, up to a maximum of 14.
It is also possible to standardize the system using samples with a known value.
The calibration curve of CDR WineLab Touch for determination of pH in wine, obtained using reference methods, confirms the excellent linearity of the system over the whole testing range.
Composition of the Kit and Reagents
Code *300310: The kit, designed for 10 tests, comprises:
1) R1: pre-filled cuvettes with reagent.
pH measures the dissociation grade of acids present in wine. This parameter influences the vinification and conservation of wine. The range of pH value is approximately 3.00 – 4.00. There is not any legal limit.
pH is determined through acid-base colorimetric reaction in which a phenol derivative, for addition of wine, undergoes a decoloration. The entity of this decoloration, measured at 630nm, is correlated to pH in the sample.
Processing - Sample Volume - Measuring Range
Wine: use as is.
(g/L of acetic acid)
|pH||3,00 - 4,00||50 µL||0,01||+/- 5|