Calcium analysis allows, in a simple and fast way, to quantify the concentration of the analyte in the water used in the brewing process
The systems to test calcium in water used in brewing process
Calcium levels in the water used in the brewing process can be measured using CDR BeerLab®, a system which enables you to carry out a wide panel of chemical analyses on beer and water, using a single instrument, in a process that is faster and simpler than traditional methods.
With CDR BeerLab® brewers and breweries of any size can independently carry out brewing process controls without having to rely on external laboratories.
|CDR BEERLAB®||CDR BEERLAB® JUNIOR|
Beer, Wort and Water:
Alcohol by Volume (ABV)
Vicinal Diketons (VDKs)
Total sulfur dioxide
Free Amino Nitrogen by OPA (FAN)
Lactic Acid (D + L)
Polyphenols in beer
Alkalinity of water
Bicarbonate in Water
Calcium in Water
Zinc in Water
Sulfate in Water
Magnesium in Water
Potassium in Water
Chloride in Water
Configuration with tailored panel of analyses
|Simultaneous Sample Analyses||16||3|
|Multiple Analyses on a Sample||–|
Test type: End Point.
Time testing: one test 1 minutes.
Are possible test sessions with several samples, up to 16 for CDR BeerLab® and 3 for Junior CDR BeerLab®
The calibration curve was constructed using standard calcium solutions. The trend of the curve confirmed an excellent linearity of the system over the whole measuring range.
Calcium reacts with a chromogen and generates a colored complex, whose absorbance measured at 580 nm is proportional to the amount of the analyte present in the water sample.
Reagent test Kits
The reagents are packaged in foil pouches containing 10 tubes useful to perform 10 analyses.
This is the minimum package that allows the use of CDR systems even to those who need to make a few analyses, thus not wasting reagents.
There are also boxes of 100 tests, however, packaged in 10 bags of 10 tubes containing the reagent.
- Codice *300695 – The kit, designed for 10 tests
- Codice *300700 – The kit, designed for 100 tests
The reagents have a shelf-life of 12 months.
Use water as is.
|Calcium||20 – 250 ppm||0.1 ppm||0.5 ppm|